Biplane cardiovascular X-ray system - risk of interruption in treatment/procedure
- Medicines and Healthcare products Regulatory Agency
- 17 December 2014
- 24 June 2013
- Alert type:
- Medical device alert
- Medical specialism:
- Cardiology, Radiology, and Vascular and cardiac surgery
(Philips) Table will reboot if users attempt a powered and manual movement at the same time. (MDA/2013/045)
Action underway: 8 July 2013, action complete: 24 July 2013
Note: These deadlines are for systems to be in place to take actions.
Biplane cardiovascular X-ray system:
Allura Xper FD10/10.
Manufactured by Philips.
All systems are affected.
Risk of interruption in treatment/procedure due to an inability to move the table.
The table will reboot if users attempt a powered and manual movement at the same time.
The reboot will take approximately 90 seconds during which time no motorised movement will be possible. This has led to delay during emergency treatment. Basic imaging will be available.
- Identify affected devices.
- Inform users of the issue.
- Superintendent diagnostic radiographers
This MDA has been sent to:
- Care Quality Commission (CQC) (headquarters) for information
- Clinical commissioning groups (CCGs)
- HSC trusts in Northern Ireland (chief executives)
- Local authorities in Scotland (equipment coordinators)
- NHS boards and trusts in Wales (chief executives)
- NHS boards in Scotland (equipment coordinators)
- NHS England area teams for information
- NHS trusts in England (chief executives)
Please bring this notice to the attention of relevant employees in your establishment.
Below is a suggested list of recipients.
CAS and SABS (NI) liaison officers for onward distribution to all relevant staff including:
- Cardiology departments
- Cardiology nurses
- Cardiology, directors of
- Cardiothoracic departments
- Cardiothoracic surgeons
- Cardiothoracic surgery directors
- Coronary care departments
- Coronary care nurses
- Day surgery units
- Health and safety managers
- Medical directors
- Medical physics departments
- Radiographer superintendents
- Radiology departments
- Radiology directors
- Risk managers
- Vascular surgeons
Establishments registered with the Care Quality Commission (CQC) (England only)
This alert should be read by:
- Hospitals in the independent sector
- Independent treatment centres
Please note: CQC and OFSTED do not distribute these alerts. Independent healthcare providers and social care providers can sign up to receive MDAs directly from the Department of Health’s Central Alerting System (CAS) by sending an email to: firstname.lastname@example.org and requesting this facility.
Philips Customer Care Service Centre
Guildford Business Park
Tel: 0870 532 9741
If you have any comments or feedback on this Medical Device Alert please email us at: email@example.com
If you are in England, please send enquiries about this notice to the MHRA, quoting reference number MDA/2013/045 or 2013/001/024/401/008.
David Grainger or Ann Seeruttun
Medicines & Healthcare products Regulatory Agency
151 Buckingham Palace Road
London SW1W 9SZ
Tel: 020 3080 7199 or 7161
Fax: 020 8754 3965
For clinical aspects please contact the MHRA clinical team on 020 3080 7248 or 020 3080 7032.
How to report adverse incidents
Please report via our website: Reporting adverse incidents involving medical devices
Further information about CAS can be found on the CAS website
Alerts in Northern Ireland will continue to be distributed via the NI SABS system.
Enquiries and adverse incident reports in Northern Ireland should be addressed to:
Northern Ireland Adverse Incident Centre
Health Estates Investment Group
Dundonald BT4 3SQ
Tel: 02890 523 704
Fax: 02890 523 900
How to report adverse incidents in Northern Ireland
Enquiries and adverse incident reports in Scotland should be addressed to:
Incident Reporting and Investigation Centre
Health Facilities Scotland
NHS National Services Scotland
1 South Gyle Crescent
Edinburgh EH12 9EB
Tel: 0131 275 7575
Fax: 0131 314 0722
Enquiries in Wales should be addressed to:
Improving Patient Safety Team
Cardiff CF10 3NQ
Tel: 029 2082 3922
Published: 17 December 2014
Issued: 24 June 2013
Alert type: Medical device alert